Measurement counter can be useful for control of changing sensor.
Long life sensor AS5000 (with certificate of conformance) is equipped as standard.
Automatic measurement to standardize the process
Minimize individual differences in how to complete measurement and read result.
Automatic temperature measurement：AUTO HOLD Function
The measurement will end automatically when tip temperature is stabilized. The measurement process will be the same no matter who does it.
Constant display of measured max. temperature：MAX HOLD Function
Max measured tip temperature will be displayed. This function can be useful for quality control of components and P.W.B.
Temperature Sending Function
Measurement result by AUTO HOLD function or MAX HOLD function can be sent to a soldering station (FN-1010) through infrared. This function can eliminate manual process of recording results, the time for the process, and human error.
Temperature Measurement Count Function
Number of tip temperature measurement will be automatically counted. This function can be useful for control of changing sensor.
Auto Shutoff Function
The power will automatically be shut off after 3 minutes of no use. It can save battery power and time for changing battery.
Fine sensor can deliver accurate measurements.
Suitable for Lead-Free Solder
Disposable sensor always maintains accurate temperature measurment
(Replace sensor when the measuring point underwent degradation)
Model No. FG-100B
Power supply: 006P 9V dry battery (alkaline cell recommended)
Temperature resolution: 1ºC
Temperature measurement range : 0 to 700ºC
Temperature precision: ±3ºC (300 to 600ºC) ±5ºC (other than above)
Temperature sensor: K (CA)type thermocouple
Display: LCD : 3 1/2 digits
Operating environment: Ambient temperature/Humidity range : 0 to 40ºC, max.80% RH (without condensation)
Packing List: Unit, 006P 9 V dry battery (for trial), Sensor (qty 10 ), Instruction manual
Replacement Parts: AS5000 Sensor
The sensor replacement period is about 50 times per single sensor in general, but this changes depending on the temperature used and the components in the solder and flux. If you use lead-free solder, the sensor needs to be replaced more frequently. If the sensor degrades because it is being eaten away by the solder, it will register a low temperature. Even if the sensor is not broken, replace it after about 50 measurements. Even if the soldering tip has degraded, the temperature will register extremely low. Keep these points in mind and decide on the sensor replacement period. Caution: The sensor is extremely thin and may break if pushed with force.